Hotplate 8 Magnetic Stirrer • Place vessel on the work plate• Set the target stirring speed and begin• Observe the stirring bar and LCD display digital model• Set the target temperature and start heating• • Observe the real temperature on LCD display digital model• Stop the heating and stirring functions.
Put the red cable inside the little socket this is your positivePut the black cable on the outside of the plug this is your negativeGrab your tape and tape these connectors nice and tight so it doesnt shift during useAfter the connecters are taped give it a little tug to make sure theyll stay in place.
A magnetic stirrer is a device widely used in laboratories and consists of a rotating magnet or a stationary electromagnet that creates a rotating magnetic fieldThis device is used to make a stir bar immerse in a liquid quickly spin or stirring or mixing a solution for exampleA magnetic stirring system generally includes a coupled heating system for heating the liquid Figure 6.
A magnetic stirrer is a laboratory instrument used for mixing liquidsIt is mainly used for stirringor heating and stirring stirring low viscosity liquid or solid liquid mixture at the same time.
A timeout of the timer automatically turns off the heater and the stirrerThis provides a convenient way to preset the length of time a sample is to be heated without the need for anyone to be present at the end of the periodGLASSCO 1 Digital Hot Plate Magnetic Stirrer OPERATING MANUAL 610.
Apr 24 2021nbsp018332A magnetic stirrer is a handy item to have at hand for a range of applications including agitation dispersion and dissolutionProfessionallymanufactured magnetic stirrers usually aren’t very pricey but if you need multiple the costs soon add upIn this set of instructions the stir bar is made from two magnets housed in moldable.
Do not place hot stirring vessels on top of the magnetic stirrer 2Maximum temperature 56176C except MICRO maxCAUTION TELEMODUL 20 C 40 C If you selected the shake mode vertical mixing Select the minimum speed setting and then increase it slowly.
Drill 4 holes just large enough for the 4 screws to go inTake the hard drive magnet and try to center it as best as you can on the fanYoull want the magnet facing away from the fanUse superglue to attach the metal side of the magnet to the fanDo not otherwise modify the fan.
For instructions on how to return endoflife equipment producersupplied electrical accessories and all auxillary items for proper disposal.
Safety Instructions Warning • Read the operating instructions carefully before use• • Temperature must always be set to at least Ensure that only trained staff works with the instrumentThe 34 Risk of burn • Caution when touch the housing parts and hotplate which can reach temperature of 0176C.
Magnetic Stirrer Instructions • Use a BNC cable to control the Magnetic Stirrer from the Fluorometer Control UnitPlug the cable in the Fluorometer AUX6 output and in the Magnetic Stirrer as shown on the picture.
Magnetic Stirrer operating instructions 1Magnetic stirrer is powered from 220V110V power sourceConnect the power cable to corresponding connector and plug it into power socketUse the BNC cable to connect the magnetic stirrer to Fluorometer Measuring unitThe stirrer is controled via the Fluorometer when it is connected.
Maintaining an appropriate agitation is important for achieving optimal results with a number of our plating solutionsThis magnetic stirrer is ideal as it takes up minimum space on your benchtop and can manage our 1 litre plating beakers with easeIt can also reduce the cost of using our beaker plating system as it can replace our more expensive hotplate stirrer where your plating solution can be operated at room temperature20 °C For guidance on rate of agitation please see the Tech Specs tab on the plating solution product page.
Manuel d’instructions 11 Manual de Instrucciones 20 Bedienungsanleitung 29 Instruction Manual Magnetic Induction Stirrer2 Safety Precautions The Magnetic Induction Stirrer is a solidstate electromagnetic coil device designed for laboratories in the bioscience medical and chemistry fieldsThis instruction manual cannot.
Operating instructions Magnetic stirrer 08001 08002 08004 BA00800003EN Albert Handtmann Armaturenfabrik GmbH amp CoDelivery Completeness and Storage 2 4.
852 Digital Display Constant Temperature Magnetic Stirrer can mix liquid precisely stably in wide speed and temperature range especially fit for mixing small volume sample.
Select Magnetic Stirrer or Overhead StirrerSelect your device Magnetic stirrer Overhead stirrer The Process Parameters window is displayedEnter the parameters and click Create or Skip this step see section Process Parameters.
So the magnetic stirrer took shape pretty quicklyI would feed the fan motor 24V through the pulse width modulation speed control board which would give it more power and more speed to handle big stirring jobsI added a switch so that the motor would run from 12V for lighter loads or from 24V for higher loads.
Switch on the magnetic stirrer 8211 the green light above the plus and minus speed control buttons will come on.
The HotplateStirrerHotplateStirrer is supplied with a power cord that is inserted into the IEC connector on the back of the unit first then it can be plugged into a properly grounded outletThe 120V unit plugs into a 120 volt 5060 Hz sourceThe 230V unit plugs into a 230 volt 5060 Hz source.
• Follow the safety instructions guidelines occupational health and safety and accident prevention regulations• CautionMagnetism Effects of the magnetic field haveMagnetic stirrer Plugin power supply Operating instruction Correct Use • Use For mixing liquids • Range of use Laboratories Schools Pharmacies 4.
• It is recommended to switch the Stirrer off during the experimentsThe noise of measured signal is lowered in this way• The frequency of stirring can be adjusted manually by the black button on the front panel of the Magnetic Stirrer• It is advisable to use a magnetic bar of 8mm x 3mm for a standard cuvette.
Copyright © 2021 Zilong Laboratory Machinery Company All rights reserved sitemap